Name: Purge Sinter - Sintering Furnace Decontaminator for Zirconia
Why should you purge your dental sintering oven?
1- Contamination builds up from oxides in coloring agents used to shade zirconia crowns or bridges.
2- These oxides cause greening or yellowing of the zirconia porcelain during sintering.
3- The oxides also build up on the furnace coils and will alter temperature reading.
Purge SINTER used regularly will eliminate all of the above.
Purge SINTER should be used 1X week for 10 minutes @ 800 degree C
Fill the provided crucible with Purge SINTER
Place the crucible containing Purge SINTER in your sintering furnace
Low Temp: 500 C / 900 F
High Temp: 800 C /1475F
Heat Rate: 40 C / 100 F per minute
Hold Time: 10 minutes
Cool Time: 5 minutes
DO NOT EXCEDE 800 C /1475 F
SHAKE WELL BEFORE USE
Use once a week
Download MSDS Sheet for Purge Sinter
Click to view Purge Sinter Brochure
BEFORE using Purge Sinter - with contamination AFTER using Purge Sinter - NO contamination
Click on photos for larger view
Dental ceramic ovens, dental furnaces, dental ovens or dental sintering ovens are used in the Dental milling laboratories. In Dental CAD/CAM restorations, these dental furnaces can employ firing, sintering and pressing functions. Dental ceramic ovens can be sintering furnaces or press furnaces. Cleaning the sintering furnaces in the Dental Milling Centers is essential to achieving the zirconia porcelain shade without greening or yellowing caused by the oxides used in the coloring agents to shade the Zirconia . Purge Sinter brings new technology for removing contamination from the dental sintering oven in the dental laboratory.
Click here for article about Purge Sinter